Biotube (Biotube) - 01.09.14
Science Objectives for Everyone
The Biotube is a single middeck locker replacement facility designed to provide a large triple-containment experiment volume with a command and data management system for experiments onboard the International Space Station.
Science Results for Everyone
Bionetics Corporation, Cape Canaveral, FL, United States
Kennedy Space Center, Space Life Sciences Laboratory, Cape Canaveral, FL, United States
Sponsoring Space Agency
National Aeronautics and Space Administration (NASA)
Human Exploration and Operations Mission Directorate (HEOMD)
ISS Expedition Duration
September 2000 - November 2000
Previous ISS Missions
The Biotube was used on STS-107 for the Magnetic Field Apparatus experiment.
- Biotube is a fully automatic apparatus that does not require continuous supervision. It allows on-orbit crew interaction through the front-panel user interface and provides for environment (temperature, humidity, pressure) monitoring.
- It also delivers fixative to the specimen and provides a digital imaging system that allows video images to be viewed on the front-panel user interface and downlinked in real time during the mission.
The main hardware components and their functions are summarized as follows:
- Generic External Shell - Provides generic housing for Space Shuttle Program middeck experiments. It houses all the subelements of the Biotube experiment, providing structural support to the subelements via a pair of attached rails.
- Generic Containment Unit - Provides three levels of containment to all subelements of the Biotube except the CDMS. This is necessary, because the hardware contains water and a fixative liquid that must be restrained should the Biotube malfunction.
- Water Delivery System - Consists of the micro-effusion delivery unit for space applications, a stepper motor, and tubing. It is controlled by the Biotube command system, which can activate the system and control the delivery of liquids without the intervention of the crew . The system can deliver up to 0.6 mL of water to each sample container. No fixative is present in the Water Delivery System. A piston-driven system ensures that the water is delivered simultaneously to all sample containers.
- Fixative Storage and Delivery System - Stores and delivers the fixative to the sample containers. This is a self-contained and sealed system. An air pump pressurizes the fixative storage container, triggering the opening of a valve to allow the fixative to flow serially through all sample containers. A waste bag at the end of the circuit takes in the fixative as it is expelled from the sample containers.
- Digital Imaging System - The system consists of 4 charge-coupled device (CCD) cameras, a video multiplexer, a lighting multiplexer, and 16 infrared light-emitting diodes (LEDs). The LEDs provide illumination while the photos are being taken.
- Command and Data Management System - This system houses the bulk of the electronics used to provide overall control of the integrated subsystems. It also houses the man-machine interface (the latter consists of a graphical user interface with a mouse pointer, system status and error message display, image display, and other functions).
- The Biotube is an autonomous system controlled by an embedded control system. It allows crew interaction on-orbit through the front panel user interface.
Ground Based Results Publications
- Life Science Services Contract
- NIH BioMed-ISS Meeting Video Presentation, 2009?Biotube
- NIH BioMed-ISS Meeting, 2009?Biotube
- NASA News and Features
Biotube Generic Containment Unit with cover removed. Image courtesy of Kennedy Space Center.
+ View Larger Image