Establishment of dosimetric technique in the International Space Station (ISS) with Position Sensitive Tissue Equivalent Proportional Chamber (PS-TEPC) - 02.11.15
The Position Sensitive Tissue Equivalent Proportional Chamber (PS-TEPC) is a radiation measuring instrument that measures absorbed doses and path length of space radiation particles simultaneously, and determines the real time Liner Energy Transfer (LET), and equivalent doses, to assess radiation risk to crew members during space flight. Dose management for space radiation exposure is extremely important for crew members, because the dose rates resulting from radiation sources (galactic cosmic rays, solar particle events, and protons trapped in Earth’s radiation belts) and secondary particles (such as neutrons) in Low Earth Orbit (LEO) are a few hundred times greater than those on the ground. The detector of PS-TEPC is a time projection chamber (TPC) made of biological tissue-equivalent materials, where a micro-pixel chamber (µ-PIC) is used as a 2-dimensional position sensor, which can acquire 3-dimensional tracks of each particle, as well as its energy, that contributes to dose distributions occurring in the human body. Science Results for Everyone
Information Pending Experiment Details
Japan Aerospace Exploration Agency (JAXA), Tsukuba, Japan
Sponsoring Space Agency
Japan Aerospace Exploration Agency (JAXA)
ISS Expedition Duration
March 2015 - Ongoing
Previous ISS Missions
Information Pending ^ back to top
- Existing active dosimeters which have been developed for real time radiation monitoring so far have defects in terms of accuracy which is caused by systematic error. Since cosmic rays have a various LET (Linear Energy Transfer), it is essential to directly measure its LET (namely deposit energy and track length of radiation simultaneously) to evaluate radiation dose on human body. The ideal real-time dosimeter needs capabilities to measure deposited energy in a certain volume and its path length in it. But the dosimeters so far used can not get position information and path lengths interpolated or estimated on ground are normally used. It is rather difficult to get path length information in real-time in space. PS-TEPC has been developed as a dosimeter having fine position sensitivity as well as capability of energy measurement and made of tissue-equivalent materials.
- This experiment aims to validate the effectiveness of PS-TEPC under space radiation environment and improved accuracy as a real time radiation monitoring device.
- Real time radiation data (energy, LET, dose, etc.) will be able to be provided more accurately.
PS-TEPC is constructed as a gas-filled time projection chamber (TPC) with a micro pixel chamber (μ-PIC) electrode. The μ-PIC is made from a series of pixel-like electrodes on a double-sided printed circuit board (PCB).
Its effective volume is surrounded by the drift plane, drift cage, and μ-PIC electrode. The drift plane and shaping frames that produce a homogeneous electric field are made of A-150 tissue-equivalent conductive plastic plates.
The anode and cathode electrode strips are arranged perpendicular to each other and their pitch is 400 μm. The z coordinate is obtained from the drift velocity of ionized electrons and drift time, which are determined using the triggering time of the plastic scintillator just behind the beam windows and the arrival time of the drift electrons at the μ-PIC electrode in the heavy ion irradiation experiment.
To evaluate the radiation dose in space, Linear Energy Transfer (LET) measurement is essential. The detector was filled with an atmospheric tissue-equivalent (TE) gas, which realize to evaluate radiation effects for human body. This methodology realizes real-time measurements of deposited energy and track of radiation in the effective volume in order to determine LET for tissue- equivalent object in space.
PS-TEPC can be used for radiation dose management onboard the ISS and future manned space missions, since it can precisely measure the doses due to various charged particles covered under a wide range of the LET energy spectrum.
PS-TEPC methodology can be applied to environmental monitoring, and dose management of radiation workers, and researchers working at high-energy accelerator facilities and nuclear facilities.
This experiment require simultaneous measuring with PADLES dosimeters and should start by 4 days after arriving onboard due to PADLES dosimeter constraints that the measurement period should be more than 90% of the total exposed duration. Also this experiment should start as soon as possible, because gas contained in the detector may cause degradation of the purity which affects the result of the experiment.
Crew unstows J-TEPC from CTB, and sets up in JEM. Measurement is operated by ground. After finishing the measurement, crew removes and stows J-TEPC. Dosimeter attached on the detector unit removal tasks during operation period. Stows PADLES dosimeters into return bag.
Information Pending^ back to top